See Item 8 for further description of key operation Enter. Don’t have an account? If upon power up the Optimus has no application program loaded, then the following Application Manger menu options will appear on the display: Except where otherwise noted, content on this wiki is licensed under the following license: Click with the right button the application icon i. Right click on the dataread.
|Date Added:||5 July 2009|
|File Size:||16.56 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
When the USB cable is connected, Windows 7 might not be able to recognize the driver. This equipment generates, uses and can radiate radio frequency energy and, if not installed and used in accordance with the instructions, may cause harmful interference to radio communications.
The lithium-ion rechargeable battery provides the Optimus with more than hours of operation. Multifunctional Keypad Multifunctional Keypad Figure 5. In order to get started the unit must have a fresh battery inserted into the battery compartment.
Select Run this program as an administrator Click ok. System Settings And Operation Tests System Settings and Operation Tests Depending on the application program that is active on the Optimus, there are number of settings and options that may be selected for both setup and testing. Install the driver in the PC following the installation wizard.
Metrologic SP5500 User Manual
If no charging information shown on Does metrologc power up after the display, reload the battery and check if the battery is properly installed then try pressing POWER key. Select Receive Data Use the Windows device manager to check out the port number assigned. When trying to load SP Data collector on Windows 7 64 bit.
Although the Optimus Software Suite does not have official support for Windows 7 nor for Windows Server R2, it has been tested on ep5500 bits and 64 bits editions of Windows 7 and Windows Server R2. Right click on the dataread. If it did come from IBC, and you are unable to get the data collector functioning, please refer to Ivan’s checklist here: No part of this work may be reproduced, transmitted, or stored in any form or by any means without prior written consent, except by reviewer, who may quote brief passages in a review, or provided for in the Copyright Act of Comments to this Manuals Your Name: For the USB cable plug the USB end of the cable into an appropriate communication port on the host device and the opposite end of the cable into the communication port of the cradle.
What does the error message: Introduction Introduction Product Overview The SP Optimus S Portable Data Terminals are robust and versatile data terminals designed to provide exceptional performance, while enduring the demands of everyday use. When the system is in alpha-mode, a small icon will be shown in the lower right corner of the display.
If this is the case, proceed as follows:. Enter text from picture: Except where otherwise noted, content on this wiki is licensed under the following license: Unable to configure IR device mean?
The driver will install and assign a new COM port which can be metrologuc out using the Windows 7 device manager.
METROLOGIC SP USER MANUAL Pdf Download.
If a customized PC application has been developed for communication with the terminal via the cradle, it will be necessary to first configure the cradle through programming. However, some considerations must be taken into account.
Click on compatabilty tab 6. These limits are designed to provide reasonable metrologlc against harmful interference in metrolgic residential installation. The stand-alone version of this driver can be downloaded clicking here ftp: The table shows a number of those settings and their descriptions. If this is the case, proceed as follows: Print page 1 Print document 18 pages.
If it did come from IBC, and you are unable to get the data collector functioning, please refer to Ivan’s checklist here:.